Determination of cell death in the BioLector® II/ Pro
Applicational approaches of propidium iodide measurement
In this application note m2p-labs presents the online propidium iodide (PI) measurement during cultivation processes using the BioLector® Pro. This method promises a rapid signal response, applicable for example in high throughput cytotoxicity screenings in 48 well format.
In general, the PI staining is implemented to evaluate the cell viability of bacterial cultures from food products, clinical samples, environmental or fermentation processes or to characterize eukaryotic cells.1 In microbiological studies, bacterial viability tests are a crucial analyzation tool, widely used for the evaluation of antimicrobial properties as in high throughput cytotoxicity screenings or in toxicity tests often performed with premixed, ready for use staining kits based on detection of membrane integrity. Propidium iodide is a versatile dye, most commonly used for identifying dead cells in a population and as counterstain in multicolor fluorescent techniques. It is supposed to penetrate only dead cells with disrupted cell membranes and is generally excluded from viable cells with intact cell membranes and thus cannot be penetrated by staining compounds.2
Via intercalating with cellular deoxyribonucleic acid (DNA) PI is emitting red fluorescence. 1 It binds to DNA between the bases with little or no sequence preference with one dye molecule per four to five base pairs, similar to ethidium bromide. In aqueous solution, the dye excitation/emission maxima of the dye are 493 / 636 nm. 3 Once the dye is bound, the fluorescence excitation maximum is shifted around 30-40 nm to the red wave length range. The fluorescence emission maximum is shifted 15 nm to the blue wave length range, resulting in an excitation maximum at 535 nm and fluorescence emission maximum at 617 nm.3 as shown in figure 1. However, with the increasing proportion of dead to live cells the red PI fluorescence increases linearly.2 PI delivers a rapid, quantitative analysis and the possibility of wide use in various instruments: fluorescence microscopes, confocal laser scanning microscopes, flow cytometers or microplate readers.3
Figure 1: Fluorescence spectrum of PI bound to DNA4
Cultivation of E. coli: The cultivation of E. coli Bl21 (DE3) takes place in the BioLector® Pro at 37° C cultivation temperature and 800 rounds per minutes (rpm) shaking speed and 800 µL filling volume per well in each cultivation well of the FlowerPlate®, type BOH2 micro titer plate (MTP). For the cultivation media, Wilms-MOPS either with 50 mM or 200 mM buffer concentration is used.
PI concentration: The PI stock solution (Sigma Aldrich, 2019) is prepared by dissolving 1 mM in phosphate-buffered saline (PBS). This stock solution needs to be sterile filtrated and stored at 4° C. As PI intercalates the DNA, the final PI concentration in the cultivation media should not be unnecessarily high to reduce the chance of hazard to the user. In this application note the final PI concentration in the cultivation medium was 5 µM. For less cross-fluorescence signals, it is recommended to use colorless cultivation media.
Initiating cell death: To induce the cell death during this exemplary cultivation, either the temperature is set to 50° C for or pure methanol (MeOH) is added 7 hours after the cultivation has started.
Determination of cell death: After the induction of the cell death, the PI penetrates the cell disrupted cell membranes and binds to DNA resulting in a fluorescence shift. The signal is then detected by the PI filter module.
PI filter module: Making the determination of death cells in the BioLector® Pro a success, the PI filter module (filter ID: 432) with an excitation peak of 532 nm and an emission peak of 620 nm is mandatory. This filter module can be purchased via email@example.com.
Various examples of the use of the detection of death cells in the BioLector® Pro with the help of PI staining are shown in the following.
Background fluorescence signals: A precultivation experiment (see figure 2) shows that the microbial growth was minimally impaired by 5 µM PI compared to the non-stained culture (E. coli control) resulting in a longer lag phase of + 0.5 h visible in the scattered light as well in the dissolved oxygen (DO) signal. Furthermore, a PI background fluorescence cross signal in the nonstained culture and an increasing background PI signal in the PI treated culture is observable. This can be caused on the one hand by the fluorescence of other metabolites during the growth. On the other hand, dissolved PI is able to penetrates the cell membrane of living cells as well. Membrane disruption arises for example through cell division, cell wall synthesis or by injury during stress. Thus, an increase of the PI signal in living cell populations over the time is visible. 2 This background signal should always be considered to avoid that viable cells are detected incorrectly as dead cells. For a better comparison of the impact of the different stress factors (MeOH or heat shock), stress factors were applied at the same cultivation time (7 hours) in all experiments.
Figure 2: Pre cultivation experiment showing the impact of 5 µM PI on bacterial growth, DO and the PI signal of a bacterial cultivation in the BioLector® Pro. Batch cultivation of E. coli BL21 wild type in Wilms-MOPS (containing 10 g/L glycerol). The cultivations took place in a FlowerPlate® microtiter plate (MTP-48-BOH2) with V0 = 800 µL and n = 800 rpm at 37 °C. The results show mean values of biological triplicates with following coefficients of variation (CV): For the scattered light signal: 0.58 % for E. coli control and 0.44 % for E. coli 5 µM +PI. For the DO signal: 0.50 % for E. coli control and 0.56 % for E. coli 5 µM +PI. For the PI signal: 5.99 % for E. coli control and 3.85 % for E. coli 5µM +PI
Cytotoxicity screening: An additional application of PI staining and detection in the BioLector® Pro is the cytotoxicity screening. In the next example, an E. coli BL21 (DE3) culture is stressed by the addition of MeOH, as shown in figure 3.
Figure 3: Cytotoxicity screening using different MeOH concentrations as stress factor to E. coli culture. Batch cultivation of E. coli BL21 wild type in Wilms-MOPS (with 10 g/L glycerol), adding MeOH with 9 %v/v or 27 %v/v in the final concentration. The cultivations took place in a FlowerPlate® MTP (MTP-48-BOH2) with V0 = 800 µL and n = 800 rpm at 37 °C. The results showing mean values of biological triplicates with following CV: For the scattered light: 5.33% for E. coli control and 7.09 % for E. coli 5 µM +PI. For the DO: 0.51 % for E. coli control and 1.24 % for E. coli 5 µM +PI. For the PI signal: 4.27 % for E. coli control and 4.07 % for E. coli 5 µM +PI.
After 7 hours of the cultivation time, the addition of 27 %v/v MeOH stalled growth, detectable in the scattered light signal and in the DO: The DO signal increased to the initial value (100 %) immediately after the MeOH addition, suggesting the initiation of cell death. The addition of 9 %v/v MeOH led to a slowdown in the scattered light signal indicating a decrease in the slope of the growth correlated to the DO signal. For example, in figure 2 control cells reach O2 limitation due to rapid growth rates as compared to the stressed cells with 9 %v/v in figure 3. The final PI signal after 18 hours is 2,54 a.u. for 27 %v/v MeOH and 1,57 a.u. for 9 %v/v MeOH treated cells. The values are blanked against medium containing 5 µM PI.
It is important to consider that this signal is a relative value and is highly dependent on the measurement timepoint and final cell number because of the exposure time of the cells to the PI and thus the above-mentioned background signals increasing with the time. Thus, the MeOH treatment and the PI determination should always be done at the same time point.
Determination of cell lysis: The PI measurement in the BioLector® Pro can be a helpful tool for the determination of cell lysis during the cultivation experiment. In the following example cell lysis is induced by increasing the cultivation temperature from 37° C to 50° C after 7 hours as shown in figure 4. The scattered light signal shows no further growth for E. coli with 5 µM PI and as well for the control strain one hour after the heat treatment was started. The growth curve of E. coli (+ 5 µM PI) without heat treatment is showing an exponential growth until the stationary phase is initiated after 11 hours. With a lag time of 2 hours after heat treatment the cells start to lyse and thus the PI signal increased linearly and saturated at around 14 hours of the cultivation time. The delay time of 2 hours is caused by the time the heat needs to distribute in the fluid and act in the cells. Regarding the PI curve of the non-heat-treated strain a background fluorescence signal is visible during the exponential growth phase.
Figure 4: Determination of cell lysis in the BioLector® Pro using PI staining. Batch cultivation of E. coli BL21 wild type in Wilms-MOPS (containing 10 g/L glycerol). The cultivations took place in a FlowerPlate® MTP (MTP-48-BOH2) with V0= 800 µL and n = 800 rpm at 37 °C. The results showing mean values of biological triplicates with following CV: For the scattered light signal: 1.86 % for E. coli control, 1.49 % for E. coli 5 µM+PI and 0.44 % for E. coli 5 µM+PI (no heat treatment) and for the PI signal: 6.75 % for E. coli control and 3.96 % for E. coli 5 µM +PI and 3.85 % for E. coli 5 µM+PI (no heat treatment).
Determination of stressed cells. The combination of the scattered light and the PI signal is useful to identify stressed cells in a running cultivation experiment. In the following example, E. coli BL21 was cultivated in Wilms-MOPS media using two different MOPS buffer concentrations. In figure 5 the experiment results regarding the scattered light, DO, pH and PI are plotted against the cultivation time.
Figure 5: Determination of stressed cells using PI measurement in the BioLector® Pro. Batch cultivation of E. coli BL21 wild type in Wilms-MOPS (with 10 g/L glycerol) either buffered with 50 mM MOPS or 200 mM MOPS, took place in a FlowerPlate® MTP (MTP-48-BOH2) with V0= 800 µL and n = 800 rpm at 37 °C. The results showing mean values of biological triplicates with following CV: For the scattered light: 0.57 % for E. coli 5 µM+PI (50 mM buffered), 0.45 % for E. coli 5 µM+PI (200 mM buffered), for the pH: 0.17 % for E. coli 5 µM+PI (50 mM buffered) and 0.26 % for E. coli 5 µM+PI (200 mM buffered), for the DO: 1.08 % for E. coli 5 µM+PI (50 mM buffered) and 2.14 % for E. coli 5 µM+PI (200 mM buffered) and for the PI signal: 2.93 % for E. coli 5 µM+PI (50 mM buffered) and 3.66 % for E. coli 5 µM+PI (200 mM buffered).
The solid blue line represents the results of the cultivation in 200 mM buffered Wilms-MOPS and the dashed line of the 50 mM buffered medium. If the focus would be on the biomass signal alone, it can be assumed that the biomass increases to higher values in 50 mM buffered media than in the 200 mM buffered one. However, the PI and DO signals are showing contrary results. Thus, is not due to biomass but particles released into the media due to cell disruption. The pH value of the lower buffered culture drops rapidly below pH 5.5 and thus below the optimal pH range during the exponential growth phase between 5 and 9 hours. This leads to a growth stop validated by the DO signal which shoots up after 9 hours as soon asthe pH drops below pH 5.5. Consequently, the DO jumps to the initial value of 100 %. In contrast, the pH in the 200 mM buffered media is more stable. During the exponential growth phase, it drops from pH 7.3 to 6.5 after 9.5 hours. Regarding the DO signal a presumably diauxic shift is visible at 10 hours cultivation time. Here E. coli is consuming the produced acetate resulting a slower increase of the DO towards the cultivation end. Furthermore, the pH increases to pH 6.7. Altogether, the pH optimum can be ensured during the entire batch cultivation process leading to a better condition of the bacteria in the 200 mM buffered medium compared to the lower buffered ones. The PI signal is showing a much higher signal of the lower buffered culture compared to 200 mM buffer concentration. Compared to the high buffered culture the PI signal of the lower one is increasing exponentially after 11 hours of the cultivation time due to cell disruption cause by the low pH as stress factor. In the end of the cultivation process this value is 2.5 times higher than the background signal of the 200 mM buffered culture. With the information of the scattered light signal in combination with the PI signal, it can be concluded that the growth in 50 mM buffered media isn’t better. Nevertheless, it is the cell lysis process which we can trace at the end of the culture. This lysis leads to an increase of fragments in the cultivation broth and thus to an increase of the scattered light signal and also an increase in the PI signal, due to the available free DNA.
This application note shows extensive application examples using PI staining method in the BioLector® Pro to identify dead or stressed cells during running cultivation processes. There are still many development possibilities such as the calibration of the PI raw signal with the cell count of dead cells. Furthermore, live/death viability testing could be developed in the future using additional filter modules to measure the living cell fraction, which allows ratio metric measurement of the dead/living cells in relation to the total cell count.
 Krämer, C., et al.: Time-resolved, singlecell analysis of induced and programmed cell death via non-invasive propidium iodide and counterstain perfusion. Scientific Reports, 2016.
 Stiefel, P., et al.: Q.: Critical aspects of using bacterial cell viability assays with the fluorophores SYTO9 and propidium iodide. BioMed Central, 2015.  Propidium Iodide Nucleic Acid Stain, https://assets.thermofisher.com/TFSAssets/LSG/manuals/mp01304.pdf (05.09.2018), Molecular Probes™, 2006.
 https://www.thermofisher.com/order/ catalog/product/V13241?SID=srch-srpV13241, ThermoFischer Scientific, 2019.